ML-OI-AX-HART – Option board
Multi-Drop HART input board
The ML-417 data logger can be equipped with an ML-OI-AX-HART input board to be able to connect HART compatible sensors/indicators.
The HART Communication Protocol (Highway Addressable Remote Transducer) is a hybrid analog+digital industrial automation open protocol. Its most notable advantage is that it communicates over legacy 4–20 mA analog instrumentation current loops, sharing the pair of wires used by the analog-only host systems. Although its dating back to the mid-1980s, the HART protocol is still one of the most popular protocols used in process and instrumentation systems.
The ML-OI-AX-HART board can be used in both point-to-point (single sensor) as in multi-drop (multiple sensors) mode. The board has a 12V excitation and an internal load resistor in the ground loop, it requires just two wires to both power and communicate with a sensor.
Before you get started
Some useful installation notes
The 12V wire should be connected to the “Loop +” input of a sensor and the returning wire between the “Loop -” output of a sensor and the “Loop +” input of the ML-OI-AX-HART board. In case of connecting multiple sensors (multi-drop), just connect all sensors in parallel.
The input board supports up to 8 channels, where each channel can be linked to the address of a connected sensor and to one of the values given by it (the current loop, primary, secondary, tertiary or quaternary value). The board has a configuration feature to scan for connected sensor ID’s.
Note: When a sensor is powered from the board’s or ML-417 power switch, don’t forget to specify a sufficient power-up time, which can be tens of seconds depending on the used sensor.
- Protocol: HART
- Topology: Point-to-Point & Multi-Drop
- Power switch: 12V
- Load resistor: Internal (270 Ohm)
- Input channels: 8
- Sensors: Up to 8
- Values: Current loop, primary, secondary, tertiary or quaternary
Discover the possibilities
Want to know more about what YDOC can do to help you with your quest for the right data logger? Do not hesitate to get in contact.